- Double articulating headrest with seesaw movement
- Soft cervical support
- Dental unit having overhead delivery system
- Two -3 way syringes (Tip autoclavable, with spare tips) one on unit side and other on the assistant side
- Two high speed Air rotor terminals with two rotor hand
pieces and accessories and one terminal for fiber optic. One for air
motor/micro motor having straight and contra angle hand pieces and other for air rotor terminal with two air rotor hand pieces with two spare cartridges
- LED light cure unit with minimum intensity 1200 mW/cm2
- Infection control system with non-retraction valves (Bio system /equivalent)
- All hand pieces / terminals should be kept on Autoclavable pads
- Arm of unit pneumatically locked
- All air tubing of the delivery system can be disinfected internally after every dental procedure
- One in built piezo ultrasonic scalar (max frequency 36 KHZ)
- Removable auxiliary tray (autoclavable) It should have integrated latest foot operated LED light (30000 – 50000 Lux).
- Rotatable water system with removable spittoon
- Medium Vacuum Suction and high suction (Motorized Suction)
having multiple programmes; Two programmable working positions Spitting and last working position with light ON and OFF automatically. Return to Zero position with light OFF automatically.
- Emergency stop control with luminous indication
- Programmable bowl water and cup filler water
- LED based X-ray viewer (For I.P.G/O.P.G films).Provided with right arm.
- Multi functional foot control base
- Provided with two stool with adjustable backrest tilt including an adjustable ring for foot rest
- Oil free medical grade compressor of 1. HP