High Level Disinfectant For Reprocessing Heat Sensitive Semi-critical Medical Devices.
CidalTek OPA disinfects heat-sensitive semi-critical medical devices which are unable to process for sterilization. CidalTek OPA can be used manually or in an Automated Endoscope Re-processor (AER).
Blood, other body fluids, and lubricants must be thoroughly cleaned from the surfaces and lumens of medical devices before reprocessing in the disinfectant. Use a cleaning protocol or standard such as US ASTM F1518. Blood and other body fluids should be disposed eff.
Refer to the reusable device manufacturer’s labelling for interaction on disassembly, decontamination, cleaning and leak testing of their equipment. Thoroughly rinse and rough dry all surfaces and lumens of cleaned devices before immersing in Medntek OPA Solution.
- HIGH LEVEL DISINFECTION
Immerse device completely in Medntek OPA Solution, filling all lumens and eliminating air pockets, for a minimum of 5 minutes at 20° C (68° F) or higher to destroy all pathogenic microorganisms, including Mycobacterium bovis, Pseudomonas aeruginosa, pathogenic fungi, and viruses. Remove device from the solution and rinse thoroughly with sterile water or with potable tap water.
The type of rinse water is dependent upon the intended use of the device. Please refer package insert for additional instructions and information regarding high level disinfection.
Note: - Product should be used in accordance with label instruction
Pharmaceutical, Hospital, Medical colleges, Laboratories where heat sensitive semi-critical medical devices which are unable to process for sterilization.
Bactericidal, Fungicidal, Yeasticidal, Virucidal ,Sporicidal, Tuberculocidal , Pseudomonacidal.